For you people who have one of these. I am having problem calibrating the conductivity on my HI98130. The meter is about a year old. When I place it in calibration mode and place the meter in the solution it just sits there blinking. I had no problem calibrating the PH. Does the conductivity calibrating solution go bad or do I have a problem with my meter?
Thank In Advance